E-Echo will be cotinuing to be available after this year! A new company is taking over.
Effective Regurgitant Orifice by Continuity Equation
Calcs>Tricuspid Regurgitation>ERO by CE>1
LVOT-CSA = 0.785 * LVOT-Diam2
TV-CSA = 0.785 * TV-Diam2
LVOT-SV = LVOT-CSA * LVOT-VTI
TV-SV = TV-CSA * TV-VTI
RV = TV-SV - LVOT-SV
RF = RV / TV-SV
ERO = RV / RJVTI
to get a ERO in Tricuspid Regurgitation.
1: Measure the Diameter of the LVOT. The diameter
can be of the LVOT should be obtained from the midesophageal
AVLAX view, in mid systole, and under the zoom mode.
The view of the LVOT should not be an obtuse angle
view. The aortic valve leaflets (NCC and RCC)
should be clearly seen. The zoom mode will decrease
the error present in the calculation. Since
the LVOT diameter changes throughout systole, the mid systole
measurement should be obtained.
2: Obtain a VTI of the LVOT. The
VIT of the LVOT is obtained from a PWD of the LVOT.
The PWD of the LVOT should be obtained at the
same location as the LVOT diameter measurement.
The PWD beam should not be off more than 15Â° from
parallel to the LVOT flow. After tracing the
flow on the echocardiographic machine, a VTI of the
PWD tracing will be calculated. A continuous
wave doppler (CWD) of the LVOT could be obtained,
but, since the CWD introduces ambiguity into the equation
and since flows in the LVOT rarely exceed the aliasing
velocity of the pulse wave doppler, PWD is the best
measurement mode. However, in aortic regurgitation,
there is excessive flow, so aliasing may occur.
Therefore, CWD may be required to be able to trace
the whole flow profile.
3: Obtain a Tricuspid Valve Diameter. The area
of the tricuspid valve can be calculated from this
measurement. This may enter some inaccuracy
into the equation since the tricuspid valve is not
a circle (it is oval).
Step 4: Obtain
a Tricuspid Valve VTI. Using the PWD
mode, trace the envelope of the entire (E wave and
A wave) flow profile of the tricuspid valve. The
diastolic flow profile is the TV VTI whereas the systolic
flow profile is the regurgitant jet VTI (RJVTI).
This activity has been planned and implemented in accordance with the accreditation requirements and policies of the Accreditation Council for Continuing Medical Education through the joint providership of University of Nebraska Medical Center, Center for Continuing Education and JLS Interactive, LLC.
The University of Nebraska Medical Center, Center for Continuing Education is accredited by the Accreditation Council for Continuing Medical Education to provide continuing medical education for physicians.
This activity is approved by the American Society of Radiologic Technologists (ASRT) as sonography-related continuing education (CE). Credit(s) issued for successful completion of ASRT-approved CE activities are accepted by the American Registry of Diagnostic Medical Sonography, American Registry of Radiological Technologists, Cardiovascular Credentialing International and Canadian Association of Registered Diagnostic Ultrasound Professionals. Contact the applicable registry for additional information / restrictions on their acceptance of ASRT CE credit.